You must read and agree to the License Agreements—two windows contain license agreements. Perform one of the following: Note that using the thread-forming screws permanently affixes the DIN-rail clip to the device. Select Next after selecting all desired software. Allow installer to initialize and then select Next. Depending on when you install your software, it is possible that there have been updates released for your products.
|Date Added:||24 August 2004|
|File Size:||37.9 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
If not, locate the DVD and right-click and select Explore. During the installation, the individual installers for each product will be invoked and appear as seen below.
Threaded inserts are located in the USB for mounting it to a panel. Based daqx your location, we recommend that you select: All power is supplied through USB so no external power supply is necessary, and direct screw terminal or BNC connectivity simplifies signal connections.
The drivers are available as support packages for the following vendors: Note that using the thread-forming screws permanently affixes the DIN-rail clip to the device. There is a proper order for installing National Instruments software due to how the individual pieces work together.
The drivers are available as support packages for the following vendors:.
Choose a web site to get translated content where available and see local events and offers. There are also many other drivers available to download such as VISA, serial, and Select I only want to evaluate products. Perform one of the following: Under Refine by Vendercheck the vendor of your device.
Video Instruction for Students Product Page: Other Daqxm country sites are not optimized for visits from your location. To communicate with a data acquisition device, you need to install the required driver on your system.
NI-DAQmx Software – National Instruments
When launching LabVIEW for the first time, you are prompted that your software is currently in evaluation mode and can be used for free for a limited amount of time. For secure desktop use, adhere rubber non-skid feet to the underside of your device, as shown in Figure 4.
Instalo you are upgrading an existing system, NI recommends that you back up any applications, projects, or other related files before continuing. This is not required, and can also be done using the NI Update Service after installation. The installation will now begin.
Allow installer to initialize and then select Next. This page has been translated by MathWorks.
Installing and Configuring USB DAQ Devices
The Add-On Explorer displays support packages for insfall vendor. You can activate the software with your serial number after completing the installation as discussed in the subsequent tutorial, How to Activate LabVIEW Student Edition. Refer to Figure 8 for mounting dimensions.
You can modify the installation path, but it is not recommended. You must read and agree to the License Agreements—two windows contain license agreements. After rebooting your computer or adding new hardware to your system, you may see the NI Product Registration Wizard appear.
Please take a moment to register your NI products to receive:. Back to Top 3.
Directions for installing NI-DAQmx – ToF-AMS DAQ
Complete the following steps to install the panel mounting kit on your USBxx device. Select the products that you would like to install. Click here to see To view all translated materials including this page, select Country from the country navigator on the bottom onstall this page. At this point, reboot your machine before using any newly installed software.